Payment & Shipping Terms:
|Power Supply:||9v Lithium Battery||Pressure Range:||30~600mH2O|
|Least Reading:||1mm||Housing:||ABS,Metallica Optioanl|
|Color:||Orange, Blue, Gray, Black Et||Rated Voltage:||300V|
mining prospecting equipment,
electronic water level sensor
50M Customized Geological Instruments Blue Orange Water Level Measurement Tape Dip Meter
Water level meter provide an accurate measurement method of level meter, usually used for the measurement of water level in the drilling holes or water level pipe, particularly suitable for hydropower engineering underground water level or artificially inspection dam observation. This instrument can be used in construction period, but also can be used as the long-term safety monitoring project. Comply with the dam safety monitoring technology standard.
|Depth of the measurement(m)||0~30||0~50||0~100||0-150||0-200|
The measurement of the water level by the following two parts:
1.Underground material, it is composed by water level pipe and bottom cover (not include in the system, it is need purchased separately)
2.Ground receiver-ruler water level meter, it is composed by detect head, ruler cable, receiver system and cable reel.
3.Detect head: made by steel, inside is a water resistance contact point. When the contact point contact water, the receiver system will open, and when the contact point leave the water, the receiver system will close.
4.Ruler cable: it made by ruler and cable, and use plastic technology keep them together, and avoid the steel corrosion and simple the operation, make the read more convenient and accurate.
5.Receiver system: it is composed by sounder, indicate light and peak value indicator, peak value indicate is voltage indicate.
In measuring, make the cable reel rotate and push the power button, put the detect head in the water pipe, put the ruler cable in hand and make it drop slowly. When the detect head contact point contact water, the receiver system buzzer will sound. Then you can read the data of the ruler cable and can know the water level.(if the surrounding voice is loudly, maybe you can hear buzzer, so you can observe the indicate light and voltage meter)
User should notice the following two points:
1.When the detect head contact water, buzzer will sound, indicator light on, voltage meter needle rotate. This time you should slowly put the cable and look for the right place of the water level and the pipe or hole’s depth.
2.The accuracy of the indication, which depend on the judge the right place which the buzzer and indicator light on. So practice of the operation is very necessary.
If you are interested,please do not hesitate to contact us.
Sinotechdrill International Co.,Ltd
Contact person: Mr.Liu
Contact Person: liujian